Specification Of HPLC Column
Hawach HPLC column is composed of a column tube, pressure cap, clamping sleeve (sealing ring), the screen plate (filter), joint, screw and so on. The column tube is made of stainless steel. When the pressure is not more than 70 kg/cm2, thick-wall glass or quartz tube can also be used. The inner wall of the tube requires a high degree of finish.
The difference between C18, C8 and C4 HPLC Column
HPLC columns C18, C8, and C4 have commonly used columns in the application, each of them provides with special properties. C18 has a strong retention ability to nonpolar compounds. C8 and C4 has a shorter carbon chain and are more reserved for nonpolar compounds than C18 weak, so the strongly retained compounds can be replaced by C8 and C4.
In addition, because of the short-chain length, C18 and C8, C4 in the separation of different points, the sub-size of the material will have a choice of differences. Moreover, the carbon content of C 18 is almost twice that of C 8, and that of C 4 is the lowest. C8, C4 and other short-chain alkyl bonding phases are suitable for small polarity to reduce retention and shorten the time of analysis.
Importance of Performance Assessment
Whether filled or purchased by yourself, the performance of the HPLC column should be examined before use, and it should be re-examined during use or after placing it for a period of time. The column performance index includes column pressure under certain experimental conditions (sample, mobile phase, flow rate, temperature), theoretical tray height and tray number, symmetry factor, capacity factor, and selectivity factor repeatability, or separation degree. Generally speaking, the reproducibility of the volume factor and selectivity factor is less than ±5% or ±10%. When comparing the column effect, you should also pay attention to whether the out-of-column effect has changed.